Orin devices deliver saw tooth and square waveforms and has the ability to deliver these waves in several different patterns utilizing our patented NTS modulations.
Typical shape of pulses of Ulticare Maxi LT99 and Biotorus LT100 devices
- Blue trace 1: Magnetic induction B pulsescale: 500mV ~ 2mT
- Red trace 2: current induced into tissue
Not noted in scale, current depends on specific body tissue conductivity. This sample waveform is measured indirectly, using an induction probe. It is proportional to derivation (change) of B in time (dB/dT)
Program P3 waveforms – fixed frequency
Pulses are equally spaced
Frequency can be set in range of 2 to 72 Hz
(This sample approx. 72Hz)
Program P2 waveforms – random frequency
Pulses are generated as groups with randomly selected frequency, varying in seconds.
Range is fixed between 2 to 72Hz.
Program P1 waveforms – wobbling frequency
Pulses are generated as up/down swept sequence of pulses.
Range is fixed between
4 to 40Hz.
Orin devices deliver saw tooth waveforms and have the ability to deliver these waves in several different patterns including our patented NTS modulations.